A summary of serial commands for the Velleman PS3005D lab power supply:
I have just upgraded my Mastech HY1803D lab power supply with a 10-turn voltage adjustment potentiometer. In order to perform the upgrade, you would need a 6.8K potentiometer (at least the original is). The trickiest part is removing the knob from the old pot. It is glued on, use pliers, soft padding and some leverage… Read More »
I’ve just discovered an undocumented command of the Velleman PS3005D programmable lab power supply. This PSU is available as a number of clones like Korad KA3005P reviewed at EEVBlog. I don’t know much about the serial communication protocol in the other PSU variants, perhaps they are different.